AirSense™ 10 CPAP Machine
The AirSense 10 AutoSet is a premium auto-adjusting pressure therapy device that automatically adjusts pressure levels to fit your changing needs and to ensure you’re receiving the lowest pressure necessary. The AirSense 10 AutoSet is also set to deliver therapy at the most comfortable temperature and humidity levels when used with ResMed’s new HumidAir™ heated humidifier and ClimateLineAir™ heated tube.
Engineered to be intuitive and easy-to-use, you can simply plug in the AirSense 10 AutoSet device and press Start
90W power supply unit
AC input range: 100–240V, 50–60Hz 1.0–1.5A, Class II
115V, 400Hz 1.5A, Class II (nominal for aircraft use)
DC output: 24V 3.75A
Typical power consumption: 53W (57VA)
Peak power consumption: 104W (108VA)
Internally located at device outlet, analog gauge pressure type, -5 to +45 cm H2O
Internally located at device inlet, digital mass flow type, -70 to +180 L/min
Maximum single fault steady pressure
Device will shut down in the presence of a single fault if the steady state pressure exceeds:
30 cm H2O for more than 6 sec or 40 cm H2O for more than 1 sec.
Pressure level measured according to ISO 17510-1:2007 (CPAP mode):
26.6 dBA with uncertainty of 2 dBA
Physical - device and water tub
Dimensions (H x W x D):
4.57" x 10.04" x 5.91" (116 mm x 255 mm x 150 mm)
Air outlet (complies with ISO 5356-1:2004): 22 mm
Weight (device and standard water tub):
44 oz (1248 g)
Maximum heater plate:
Maximum gas temperature:
106°F ( 41°C)
Material: Polyester non woven fiber
Average arrestance: >75% for ~7 micron dust
Material: Acrylic and polypropylene fibers in a polypropylene carrier Efficiency: >98% for ~7-8 micron dust; >80% for ~0.5 micron dust
ResMed confirms that device meets the Federal Aviation Administration (FAA) requirements (RTCA/DO-160, section 21, category M) for all phases of air travel.
CDMA (USA and Canada only) 2G GSM (all regions except USA and Canada)
FCC ID: 2ACHL-AIR10CD
The AirSense 10 device complies with FCC Rules.
The AirSense 10 device should be used at a minimum distance of 0.8" (2 cm) from the body during operation. Additional information regarding the FCC Rules for this device can be found on www.resmed.com.
Operating pressure range
AutoSet, AutoSet For Her, Elite, CPAP 4 to 20 cm H2O
Maximum flow: 4 L/min
Pneumatic flow path